4850 level, room at the WYE, train passes by but otherwise a very quiet environment. Also used to store our foam panels, there should be 12-15 complete panels and several smaller pieces.
Status: tile installed, huts built. Need to finish network/power installations, need to define a place for electronics.
Install sheet: 4850c.pdf
|GPS transceiver serial||DY5A10128|
|Switch label||4850 SW 1|
|11/19/2014||Tanner installed the full station. Still need a cable for GPS timing.||Tanner|
|11/20/2014||GPS cable installed. Timing quality at 80%, not improving.||Tanner|
|01/15/2015||Moved electronics out of insulation hut and onto a table. Installed a new GPS cable with TTL to RS-232 conversion, timing quality at 100%. Replaced DC power supply with a batter and battery tender.||Tanner|
|02/02/2015||Seismometer mass positions checked remotely, centering not needed.||Tanner|
|04/02/2015||Used the Octans to align the seismometer. The seismometer was centered after the alignment.||Tanner|