Surface site at Oro Hondo. On Lab property, but need permission from Barrick to access it.
Status: Installed concrete pad, retaining wall, insulation, and the solar panel post. Need to install the solar panel, radio, define a place for electronics.
Install sheet: oro.pdf
The sensor below was originally installed at this station, but was discovered to be faulty and was replaced on 4/3/2015. The current station information is given above.
|03/30/2015||Tanner and Daniel installed the station. Had problems with getting the seismometer working well and getting the Q330 to wake up the Baler. Will watch the seismometer remotely and see if we need to replace it.||Tanner|
|04/03/2015||Tanner and Daniel replaced the Guralp with an STS-2. Seems to be working better. Also figured out how to get the Q330 to wake up the Baler (need to set the interface to Serial 2 rather than Ethernet when using the Clie).||Tanner|
|05/01/2016||Recentered seismometer remotely.||Tanner|
|06/17/2016||Recentered seismometer remotely.||Tanner|
|08/08/2016||Recentered seismometer remotely.||Tanner|